Why Use Shockwave Massager?
Recognition of Host, Handpiece and Accessories
1. Host 2. Handpiece
3. Transmitter Heads 4. Connector
5. Power Outlet 6. Power Cord
7. Alloy Flight Case 8. Power Port
9. Power Switch
10. Handpiece with Cable
11. Transmitter Heads*7
1# 8mm: acupuncture points or small finger joints
2# 15mm: small muscle tissues
3# 15mm(concave): deeper structures
4# 20mm(concave): myofascial treatment
5# 36mm (assembled in handpiece): big muscle tissues
6# Flat head: ED treatment
7# Cylinder-shaped head: ED treatment
12. Soft Tip Heads*3
You can use them to cover metal transmitters for treatment of highly sensitive regions, as at trigger points in the cervical spine muscles.
13. Silicon Rubber Cases*5
Using them to cover metal transmitters for protecting after treatment.
Firstly, insert the connector of handpiece into your host power outlet, located in the front side of the host, making sure the bump of connector is matched with the notch of the power outlet first.
Secondly, attach the power cord for your host to the power port on the back side of the host, then insert the plug of the other end into a power socket.
Thirdly, choose a suitable transmitter head for the part of body you want to treat, unscrewing the original transmitter head on the handpiece in an anticlockwise direction and screwing the new transmitter head in a clockwise direction slowly and carefully for replacement.
Then turn on the power switch on the back side of the case and find your treatment points and apply couplant to the skin beforehand.
Press the corresponding button on the host, preset energy level and frequency will be displayed, you can adjust the parameters follow your needs by pressing the P+/P- and F+/F- buttons.
Press the power button of the handpiece for treatment.
The default striking times is 2500, it will stop working automatically when reach 2500. Press the power button for start again, then stop.
Default Parameters of Each Mode
|Host Colour||Built-in alloy case||Host Material||Aluminium alloy|
|Rated Power||90W MAX||Rated Current||2A MAX|
|Set Energy Level||10-120mj||Set Frequency||1-16HZ|
|Strike Times||2500||Transmitter Heads||7|
|Device Dimension||380*318*102mm||Packing Dimension||380*318*102mm|
|Net Weight||4.010KG||Gross Weight||4.500KG|
Features and Advantages
1 x Host with alloy case
1 x Handpiece with cable
1 x Power cord
7 x Transmitter heads
3 x Soft tip heads
5 x Silicon rubber cases
Othes: The tools for maintenance of the handpiece
Similar Products Comparing
|Preset Functional Modes||Mini Size|
|Host Colour||Built-in alloy case||Black|
|Host Material||Aluminium alloy||ABS|
|Rated Power||90W MAX||90W MAX|
|Rated Current||2A MAX||2A MAX|
|Set Energy Level||10-120mj||25-105mj|
|Built-in Functional Modes||12||/|